SEARCH THE RESOURCE LIBRARY
REGISTER FOR UPCOMING TRAINING
New Online Sales TrainingSales professionals need to keep their skills sharp in order to discover new leads, nurture promising prospects and close more deals. EASA's new streaming sales videos provide relevant and insightful training.
Need an Accredited service center?Find a service center that has proven they repair electric motors in accordance with ANSI/EASA's AR100.
VIEW THE LISTLEARN MORE ABOUT THE PROGRAM
ACCREDITED? VIEW YOUR RECORDS
Registrationnow open!Register by May 21 to take advantage of the lowest rates to attend the 2021 Convention & Solutions Expo.
REGISTER TO ATTEND RESERVE A ROOM
Engage locallyEASA's international membership is divided into 10 Regions that are made up of 32 Chapters.
EXPLORE CHAPTERS & EVENTS NEAR YOU
SEARCH THE ONLINE STORE
EASA Resource GuideA handy, downloadable PDF booklet summarizing the products and services available from EASA.
CUSTOM BRANDED PRODUCTS
Helpful guidance for assessing internal calibration activities
For many service centers, specific customer requirements regarding control of measuring & test equipment will be much more specific than as listed below. Customer requirements should always be evaluated first as they are prerequisite for doing business.
When calibrations are performed internally, calibration procedures should be used that contain adequate information for the calibration of the measuring & test equipment (M&TE).
Typical contents of adequate calibration procedures are listed below.
It is common practice for many calibration procedures to require Test Accuracy Ratios (TAR) of 4:1 or better between the standard used and the M&TE being calibrated. That is, M&TE with an accuracy of 4% would be calibrated by using a standard whose accuracy is 1% or better. Additionally, it is reasonable to adjust the required accuracy of M&TE to meet the needs of the application. For example, if 10% accuracy is sufficient for an application, the standard used for calibration may have 2.5% or better accuracy. Whether manufacturer specified accuracy or an alternative accuracy is used, the conditions of the calibration will be documented through a label on the M&TE that is linked to a calibration certificate on file at the service center that references the applicable calibration procedure.
For additional help and tips regarding equipment calibration, you may find this white paper helpful: "Minimizing Calibration Costs for Measuring and Test Equipment" by Mike Howell, EASA Technical Support Specialist.